Talos F200i TEM
Explore MyScope Outreach
Video: Solid Oxide Fuel Cells
The 2016 Image Contest Grand Prize Winner
Learn how to engage in cryo-EM SPA
2017 Nobel Prize for Chemistry
Krios G3i Cryo-TEM
Webcast: How Image Accuracy Impacts Your Research
What's new in PerGeos 1.1
The 1000 °C heating stage is used to heat samples and record in-situ morphological sample changes.
Add to my Upgrades and Accessories List
Add Item and Request a Quote
Find another Accessory
When temperatures exceed 400 °C, the specimen must be covered with a metal heat shield to prevent convective heat loss and to protect the EDX detector. The temperature accuracy is +/- 15 °C. The maximum heating rate is 300 °C/min. The high chamber pressures in ESEM™ mode acts as a buffer against high outgasssing rates in case melting and subsequent evaporation occurs.
The heating stage itself is controlled through a microprocessor-controlled power supply, providing set-point temperature control and LED read-out with 15 ºC accuracy. The stage is fully controlled through the software with the possibility to define ramp/soak profiles.
A dedicated ceramic GSE detector with a special insert is also provided, as the standard GSE and insert should not be used for heating experiments.
5-mm ceramic crucibles can be ordered separately as consumables.
Training is highly recommended to be ordered with the 1000 ºC Heating Stage.
no items have been added